MEMS air flow sensor for Hydrogen fuel cell
MEMS mass flow sensor in Hydrogen fuel cell control, measuring the air going into fuel cell.
For manual, please visit our new website.
S4000 series of gas flow sensors are specially designed for the measurement of medical gas flow and process control, which based on the MEMS flow sensor core technology , and applied to the kinds of clean and dry gas flow measurement with achievements of facility digitalization, intelligence and high safety compared to the traditional industry. S4000 series have the property of wide range of flow measurement which satisfy different flow measuring requirements with high sensitivity, high reliability and stability and low cost.
S4000 series promotes to development for more intelligence and energy saving.
Gas mask and respirator,
Continuous positive airway pressure (CPAP) device,
Mass flow controller,
Environmental climate control,
Fuel cell control
Wide flow range 0-500SLPM
High precision (1.5% F.S accuracy)
Linear output and temperature compensation
Maintain long-term stability with minimal zero drift
Solid-state sensing core (no surface void or fragile membrane), anti-blocking and pressure shock
Analog output (1 to 5 V) (I2C output is also available)
Fast response time (20ms response time)
Working temperature: -25 ° C to 85 ° C
Storage temperature: -40 ° C to 90 ° C
Impact resistant 100 g
The sensor is resistant to condensation
Function Paremeters1. Customizable range between 10SLM and 500SLM2. SLM: Standard liters per minute. Standard conditions: 0 ° C and 1 atmosphere3. In addition, the two-way airflow test can be customized, and the analog output is correspondingly changed. F(min)—F(max) corresponds to 1-5V output, and 0 flow corresponds to 3V;4. Includes temperature drift and linearity errorLinear Output1. The unidirectional airflow mode: (calculation formula)Flow = [(VOUT -1 V) / 4 V] × full scale flowFor example: S4103V has a full-scale flow of 200 SLM. When the output voltage is read at 2.5V,the instantaneous flow rate is [(2.5V-1V) / 4V × 200 SLM] = 75 SLM2. Two-way airflow mode: (calculation formula)Forward flow = [(VOUT -3 V)/2 V] × full scale flowReverse flow = [(3 V- VOUT)/2 V] × full scale flowShell SizeNote 1: The connector can be matched with a connecting line according to customer requirements.